Comtest and its Representative in Poland, AP Flyer, have recently completed and commissioned a large anechoic chamber for Bialystok Science and Technology Park.
The Laboratory includes one EMC anechoic chamber for emissions tests at 1 m, 3 m and 5 m and immunity tests at 1 m and 3 m. The chamber is equipped with a large fully automatic door and state-of-the-art Comtest HT-45 polystyrene absorbers. In addition to the anechoic chamber two ancillary RF shielded rooms for control equipment are supplied as well as amplifiers.
The chamber was independently verified by Seibersdorf and its performance meets and in some cases exceeds the given guarantees.
NSA according to CISPR 16-1-4 Ed. 3.1 - 2012-07 (Clause 5.4.4) Reference Site Method (RSM), at 5 m range in a volume of 3 m diameter x 2 m height within ± 3.5 dB
NSA according to CISPR 16-1-4 Ed. 3.1 - 2012-07 (Clause 5.4.4) Reference Site Method (RSM), at 3 m range in a volume of 3 m diameter x 2 m height within ± 3.0 dB
sVSWR according to CISPR 16-1-4 Ed. 3.1 - 2012-07 (Clause 8), at 3 m range in a volume of 1.5 m diameter x 2 m height below 5.5 dB
Field uniformity according to IEC 61000-4-3 Ed. 3.2.2010, in a plane of 1.5 m x 1.5 m, from 0.8 m from the floor, meets 100% of the points between 26 MHz and 1 GHz, and 75% between 1 and 18 GHz
The new EMC test facility will be used to perform measurements according to EN 55011, EN 55014, EN 55015, EN 55022, EN 61000-4-3 & -6, DO-160-F, MIL-STD 461-E, being a very comprehensive and state-of-the-art test facility.
Comtest thanks the Customer and AP Flyer for the cooperation and the outstanding achievement and wishes success to the Bialystok Science and Technology Park for their future of testing.